FORA TN’G SpO2 Fingertip Pulse Oximeter (Bluetooth 4.0)
Roll over image to zoom in
Click to open expanded view
- Large LCD Display
- Bluetooth Indicator
- Easy to use with one button
- SpO2, Pulse Rate and Pulse Bar
- 2 AAA Batteries
- FORA Fingertip Pulse Oximeter*1
- Operating Instructions
- AAA-Size Alkaline Batteries*2
- Warranty Card
|Model No.||TN’G SpO2|
|Dimension & Weight||63(H)x37(W)x32(D) mm, 40g without batteries|
|Battery Life||Batteries can be used continuously for 8 hours (for reference only ,
it depends on different brands of AAA alkaline batteries)
|Power Source||Two 1.5V AAA alkaline batteries|
|External Output||0% to 100%|
|Accuracy||100% ~ 80% ±2%; 79% ~ 70% ±3%; others are undefined.|
|Method||Dual wavelength LED|
|Measurement Range||30 to 250bpm|
|Accuracy||±1bpm or ±1%, whichever is greater|
|Operating Conditions||50°F to 104°F (10°C to 40°C); Below 95% R.H. (non-condensing)|
|Meter Storage / Transportation Conditions||-13°F to 158°F (-25°C to 70°C); Below 95% R.H. (non-condensing)|
|Product Life Time||12 months|
|Type BF Applied part||Type BF Applied part|
|Harmonized Standard||ISO 80601-2-61:2011|
|Mode of Operation||Spot Check / Monitoring|
Q: How to operate the TN’G SpO2 device:
STEP 1. Turn on the oximeter by pressing . Do not move your finger when starting test. Do not move your body while testing.
STEP 2. Open the clamp and put one of your fingers into rubber hole of the oximeter (it is better to let your finger touch the bottom.) before releasing the clamp.
STEP 3. After detecting the pulse signal, the oximeter shows the readings of SpO2 and pulse rate on the display. The readings will be updated based on the signal received with each pulse.
STEP 4. While testing, if you press , the screen will rotate 180 degrees.
There are no reviews yet.